MiniSpin® is a registered trademark of Eppendorf AG, Germany. .. >Read this operating manual completely before using the device for the first time. Also. Get Eppendorf MiniSpin plus Manual. Get all Eppendorf manuals!. Get Eppendorf MiniSpin plus Owner manual. Get all Eppendorf manuals!.
|Published (Last):||15 April 2018|
|PDF File Size:||6.72 Mb|
|ePub File Size:||12.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
Rotor care instructions Rotor use instructions Inspection Seal replacement Lubrication Function test. Or point us to the URL eppensorf the manual is located. MiniSpin plus MiniSpin 14, rpm 13, rpm Max.
What’s missing? Tell us about it.
Powerful and user-friendly MiniSpin and MiniSpin plus are small enough so that each workstation can be equipped with a “personal” centrifuge. Eppendorf International Choose country. MiniSpin plus provides speed sufficient for molecular biology separations.
Only neutral agents eppendor be used for cleaning purposes e. Centrifuge shakes during acceleration. Open tube lids can be torn off during centrifugation and damage the rotor lid or centrifuge.
Real—life personal centrifuges do exist in the Minispin and MiniSpin plus from Eppendorf. Product Information Powerful and user-friendly MiniSpin and MiniSpin plus are small enough so that each workstation can be equipped with a “personal” centrifuge.
Brochure — Endless Possibilities. Cleaning of housing Checking for damage Checking lid latch Checking of rotor detection System test run. Tell us about it. Contact us Distributors Technical Service. No main power connection. Or point us to the URL where the manual is located. White Paper — Routine Maintenance of Centrifuges.
Entry-level centrifuges that are powerful and easy-to-use. Rotor not loaded symmetrically. Your manual failed to upload No main power connection. Instructions for use — Rotor F Controllare il fusibile del laboratorio. La centrifuga vibra nell’avviamento. With MiniSpin plus, the time can be pre-selected between 15 seconds and 99 minutes, and with MiniSpin between 15 seconds and 30 minutes.
Emergency lid release see Sec. Fasten the rotor lid and close the centrifuge lid. Load the rotor symmetrically. Non si riesce ad aprire il Manca la corrente. Be sure to close the tube lids tightly prior to centrifugation. Disconnect the centrifuge from the mains supply, remove the rotor and clean it separately. Operating manual — MiniSpin plus. Inserire il cavo rete su entrambi i lati.
Eppendorf MiniSpin plus : Manual
Il rotore continua a girare. Checking for damage Checking lid latch Checking speed sensor Verification eppendoef operating parameters acc. Service product portfolio may vary according to country.
If this takes place during the run, the display switches back after 20 seconds.
Inspect the rotor lid regularly. Emergency lid release see Sec.
Wait for the rotor to come to a standstill. Rotor not loaded symmetrically. Switch the device off immediately by pressing “STOP”. Attendere che il rotore si fermi.